|Place of Origin:||China|
|Model Number:||SARS-CoV-2 Antigen Test|
|Minimum Order Quantity:||20,000|
|Price:||USD 2.5-USD 3.5 per test|
|Delivery Time:||3-15 business day|
|Specification:||5 Tests/box||Methodology:||Latex Immunochromatography|
|Components:||Test Card, Dilution, Capillary Tube, Swab||Storage:||Room Temperature Or Refrigerated (2-30℃)|
|Validity:||18 Months||Sample Type:||Nasopharynx/oropharyngeal Swab Collection|
|Instrument:||None||Device Type:||IVD (In Vitro Diagnostic) Medical Device|
20mins rapid ag test kit,
5pcs ag saliva rapid test card,
IVD rapid ag test kit
Highly Accurate Fast Covid-19 Test Kit/SARS-CoV-2 Antigen Assay Kit by Latex Immunochromatography Method
This product can be used as a supplementary test indicator for suspected cases with negative nucleic acid test of SARS-Cov-2 or used in conjunction with nucleic acid testing in the diagnosis of suspected cases. It cannot be used as a basis for the diagnosis and exclusion of pneumonia caused by SARS-Cov-2 infection. This product can also be used for the auxiliary diagnosis of influenza A virus and influenza B virus infection.
1 Prick the aluminum foil of the sample storage tube with the dripper, and then take out the dripper and put it aside.
2 Insert the swab after sampling into the sample preservation tube, immerse it in the liquid, rotate it close to the inner wall of the test tube for 10-20 times to dissolve the sample in the sample preservation solution; squeeze the swab head along the tube wall to keep the liquid in the tube as much as possible, and discard the swab.
3 Insert the dripper into the sample preservation tube again.
1 tear the aluminum foil bag, take out the test card and lay it flat.
2 drop 3-4 drops (about 80μl) of sample preservation solution containing sample into the sample adding hole of test card.
3 the results were observed within 15-20 minutes, but no clinical results were found after 30 minutes.
POSITIVE: Two red lines appear. One red line appears in the control region(C), and one red
line in the test region(T). The shade of color may vary,but it should be considered positive whenever there is even a faint line.
NEGATIVE: Only one red line appears in the control region(C), and no line in the test region(T).
The negative result indicates that there are no Novel coroinavirus particles in the sample or the number of viral particles is below the detectable range.
INVALID: No red line appears in the control region(C). The test is invalid even if there is a line on test region(T). Insufficient sample volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the test procedure and repeat the test using a new test device. If the problem persists, discontinue using the test kit immediately and contact your local distributor.
|Sample Type||Nasopharynx/oropharyngeal swab collection|
|Device Type||IVD (In vitro diagnostic) Medical Device|
|Specification||1or 5 or 25 tests/box|
Contact Person: Harvey Huang