|Place of Origin:||China|
|Model Number:||SARS-CoV-2 Neutrailzing Antibody|
|Minimum Order Quantity:||1000|
|Delivery Time:||10~15 work days|
|Payment Terms:||L/C, T/T|
|Supply Ability:||10,000 pcs/week|
|Sample Type:||Serum/Plasma/Whole Blood||Valid Period:||15 Months|
|Reaction Time:||8 Minutes||Package:||25 Tests|
|Test Environment:||Room Temperature||Sample Size:||150μl~250μl|
250ul saliva rapid test card,
IVD 150ul Antigen saliva rapid test card,
IVD saliva rapid test card
Use In-Vitro Diagnosis for SARS-CoV-2 Neutralizing Antibody Rapid Test Kit
Specimen collection and preparation
1. Human whole blood (heparin
anticoagulation recommended), serum, or plasma (heparin anticoagulation recommended) sample.
2. Test the sample as soon as possible after it is collected. If it cannot be measured in time, it should be transported and stored as required.
3. The samples should be transported and stored at 2℃ to 8℃. Whole blood samples can be stored for 7 days at 2℃ to 8℃, and serum/plasma samples can be stored for 3 days at 2℃ to 8℃. Stable at -20 ℃ for 7 days. Refrigerated samples must be left to recover from 20℃ to 28℃ before testing,multiple freeze-thaw cycles should be avoided. The sample should be measured during the stable period, otherwise, the sample must be collected again.
4. The test result may be affected by severe jaundice, hemolysis, and lipemia samples.
Please read the instruction manual of the applicable instrument and this manual carefully before use. The reagent should be returned to room temperature before testing, and the testing process should also be performed at room temperature.
1. Testing steps
1.1 Open the kit and insert the test card into the instrument interface.
1.2 Use the Capillarity Tube to aspirate the sample, insert it into the diluent, Mix well and drop three drops of liquid into the test card sample hole. After 8 minutes, place the test card into the instrument and click "Test".
1.3 Select the appropriate sample detection mode for testing on the instrument operation screen.
1.4 After the test is completed, remove the test card from the instrument and process it in accordance with the regulations.
Interpret the test result in accordance with the interpretation criteria of the qualitative results in the manual.
|＞20%||positive||Neutralizing antibodies were detected in the sample|
|≤20%||negative||No neutralizing antibodies were detected in the sample|
Contact Person: Eric King