Click the mouse to input the patient ID, age, sex, select the suction
head type and sample type and add the sample. The equipment will
automatically carry out the calculation and display the test result.
Testing time and testing process
Different items have different reaction durations. Please refer to the
user manual of reagent. If you click the "cancel testing" button during
testing, the testing will be ended.
After finishing testing, the test result interface will automatically pop
up. You can click the "confirm" button to close the interface or click
"Print" button to print the test result.
Function description of each button:
Back: This button is used for going back to the main interface. This
button cannot be pressed during testing.
Test: This button is used for distributing tasks. By clicking this button,
the equipment will automatically carry out testing.
Cancel: If components do not act during testing, you can click the
button to cancel testing.
Card exit: Before testing, by clicking this button, the chamber door
will be opened, and the card support will exit automatically.